transformer, test leads, and input jacks — must be purchased from Keithley Instruments. Standard fuses with For analog filter (Models A/A only) . leaving the factory the A and A meet these specifications. the Series A Reference Manual for additional power derating. Keithley A Pdf User Manuals. View online or download Keithley A User Manual.
|Published (Last):||13 July 2006|
|PDF File Size:||11.98 Mb|
|ePub File Size:||11.84 Mb|
|Price:||Free* [*Free Regsitration Required]|
Calibration is locked 1 or smuX. In absence of this minimum load capacitance, overshoot and ringing may occur. Each open script file is displayed on a separate tab. If the programmed state of the line is high, the digio.
If you want to reset a specific instrument, use either the localnode. Ford pickup f v8 frameoff great truck ford pickup f v8 frameoff great truck manual 266a truck. TSP command reference As an example, to set bit B1 of the questionable status calibration summary enable register, set status.
The following figure is an example of a five-point logarithmic sweep from 1 V to 10 V. Connect To dialog box 4.
What’s missing? Tell us about it.
It does check to see if all the pulse keithlej can be achieved, and if they can, assigns the indicated tag or index to the pulse train. Bit B1 decimal value: Trigger mode values triggerMode Description digio.
LAN concepts and settings 1. Type a name to identify the connection for example, My Instrumentand then click OK. When proper connections are made, the instrument chassis is connected to power-line ground through the ground wire in the power cord. The following figure displays a single TSP-Link network with groups. In addition to the above constants, operationRegister can be set to the numeric equivalent of the bit to set.
Set this attribute to zero to bypass waiting for an event the SMU continues uninterrupted through the remote trigger model. In noisy environments, you can manually configure the instrument to match the actual line frequency.
KEITHLEY B : User manual
Issuing Icl Commands This level is used to indicate status changes; the message does not represent an error. To configure the line frequency from the front panel: To achieve such precise timing, use a static trigger configuration. Pulse train triggering BS Rev. General operation Recalling a saved setup You can recall setups from internal nonvolatile memory or a USB flash drive at any time.
When all external drives release the physical line, the rising edge is detected as an input trigger. The magnitude of the first pulse will be start amperes; the magnitude of the last pulse will be stop amperes. Sets the trigger count number of measurements as follows: Do not use 0. Configures Channel A buffer.
Keithley 2636a reference manual
It performs the commands lan. The center conductor of the cable is used for HI, and the inner shield is used for guard. For numbers larger thanonly the lower 32 bits are used. For detailed firmware installation instructions, refer to the upgrading the firmware topic in section 21 of the series a system sourcemeter instruments reference manual document number.
The instrument automatically stores the source for all scripts that are loaded on the instrument. LAN concepts and settings 8. TSP command reference Usage bufferVar. I have looked at the reference manual for the and i think i understand how the triggering works internally, but i am unclear how to implement it using the provided vis.
The reset function has no effect on flow control. The Series B stores the hardware value of the last LXI trigger packet that was sent or received as the pseudo line state. The readings are on maunal top line, and source and limit values are on the bottom line.